Lab Tests 20190925
20190925/20190925_164951_config.xml NA
20190925/20190925_165419_config.xml NA
20190925/20190925_165852_config.xml NA
20190925/20190925_170253_config.xml NOISE    Ant6+Ant4
20190925/20190925_170415_config.xml SURVEY   Ant6+Ant4
20190925/20190925_170754_config.xml SURVEY   Ant5+Ant4
20190925/20190925_171222_config.xml SURVEY   Ant8+Ant4
20190925/20190925_171829_config.xml SURVEY   NONE+Ant4 (not sure what this was for)
20190925/20190925_172651_config.xml SURVEY   Ant7+Ant4
20190925/20190925_173100_config.xml SURVEY   Ant7+Ant2
20190925/20190925_173432_config.xml CALIBR   Ant7+Ant2
20190925/20190925_174052_config.xml SURVEY   Ant7+Ant2
20190925/20190925_174204_config.xml CALIBR   Ant7+Ant2 (repeat measurement)
20190925/20190925_175040_config.xml SURVEY   Ant7+Ant1 (repeat measurement)
20190925/20190925_175337_config.xml CALIBR   Ant7+Ant1
20190925/20190925_175640_config.xml SURVEY   Ant7+Ant3
20190925/20190925_175848_config.xml CALIBR   Ant7+Ant3

Loopback setup was used for all measurements

Transmitter (1000 W) --> Cable
	--> 30 dB high power attenuator
	--> 6 dB Werlatone J1 to J2 (i.e. 1 dB atten)
	--> 15 dB attenuator
	--> 7 dB attenuator
	--> Cable
	--> 10 dB attenuator
	--> 15 dB attenuator
	--> Fiber delay line, ? attenuation
	--> Cable
	--> Splitter (Minicircuits 10-1000 MHz 75C-2-4+), 3 dB attenuation
	TO RECEIVER: TOTAL 81+fiber_delay_line dB attenuation
	Splitter Port 1 --> 30 dB attenuator
		--> Cable
		--> 10 dB attenuator
		--> 15 dB attenuator
		--> Receiver TOTAL: 136+fiber_delay_line dB attenuation
	Splitter Port 2 --> Cable
		--> 10 dB attenuator
		--> 15 dB attenuator
		--> J3 to J1 connector (i.e. 6 dB atten)
		--> 30 dB high power attenuator
		--> Cable
		--> Transmitter/Receiver TOTAL: 142+fiber_delay_line dB attenuation
	We expect 6 dB more attenuation than Receiver and measure ~7 dB more

Expected transmit power level is 60 dBm - (142+fiber_delay_line dB attenuation) = -82 dBm - fiber_delay_line dB attenuation
Expected receive power level is 60 dBm - (136+fiber_delay_line dB attenuation) = -76 dBm - fiber_delay_line dB attenuation
Measured receive power is ~-81.5 dBm

CALIBRation mode
In calibration mode
	24.96 us delay for receive channels
	24.97 us delay for transmit channels (10 ns greater time delay than receive)
Signal power on screen is 158 dB for Ant7
Receive channel power measured on DSO: 30 mVpp == -26.5 dBm (after 55 dB of attenuation)
	-26.5 - 55 = -81.5 dBm into the receiver
Signal power on screen is 151 dB for AWG3, ADC2, ANT4 (top right connector)
Signal power on screen is 151 dB for AWG1, ADC1, ANT2 (bottom right connector)
Signal power on screen is 151 dB for AWG0, ADC3, ANT1 (bottom left connector)
Signal power on screen is 151 dB for AWG2, ADC4, ANT3 (top left connector)
Signal power on screen is 158 dB for Ant7

SURVEY mode
Noise power on screen is 42-43.5 dB
Signal power on screen is 161 dB for Ant7
Signal power on screen is 161 dB for Ant6
Signal power on screen is 160 dB for Ant5
Signal power on screen is 156 dB for Ant8
Signal power on screen is 154 dB for AWG3, ADC2, ANT4 (top right connector)
Signal power on screen is 154 dB for AWG1, ADC1, ANT2 (bottom right connector)
Signal power on screen is 154 dB for AWG0, ADC3, ANT1 (bottom left connector)
Signal power on screen is 154 dB for AWG2, ADC4, ANT3 (top left connector)


Anechoic Chamber Tests 20190926
013430: PA off, noise settings, 50 ohm
014203: PA on, noise settings, 50 ohm
020409: PA on, noise settings, antennas
020822: PA on, regular settings amp 0.25, antennas
021308: PA on, regular settings amp 1, antennas
021703: PA on, noise settings, antennas
021957: PA on, regular settings amp 1, antennas
022901: PA on, regular settings amp 1, antennas
    Removed extra rg_start_offset

